Product performance: the temperature test range: 25 ℃ ~ 45 ℃;
Temperature test precision: ± 0.1 ℃;
Product structure: consists of plug, cable, sensor;
Compatible models: Philips A3 series monitor: M3926A, M3927A, M3928A, M3929A;
Ma Kui 7000/7010；GE Eagle, Solar, Tram, Tramscope, MAC - Lab Systems.
Application Scope: multi-parameter monitor;
Contraindications: this product may not be fixed in the position of the tissue damage
Service life: it is recommended that each use replaced every 6 months.If you still need to use, please evaluate, and use should be tightened during the inspection.
1) the correct selection of sensor types, if measurement location for axillary, select surface probe, if the measurement position of rectum, chose the red tag body cavity probe, if the measuring position for oral, should choose blue tag body cavity probe, rectal probe and dental probe shall not be mixed.
2) if can't accurately monitoring temperature sensor, it is the location of the wrong place safe or not, if you have any of the above, reposition the sensor or choose another type of sensor.
3) this product can not be into any cleaner, use of alcohol or alcoholic washing solution will directly affect the service life of the cable.
4) if can't accurately monitoring pulse sensor, it is placed in the right position or its parts are too thick, such as the above situation occurs, reposition the sensor or choose another type of sensor.
5) to avoid damage to the cable, disconnect with the handheld plug instead of cable.
6) do not use in MRI scans, due to current conductor can cause burns and formation of interference signal.
7) connection will lead to improper equipment data discontinuity or don't show;
8) this product sterilization disinfection way is to use epoxy ethane.
9) shell is damaged or cable sheath after damage do not use.
Directions for use:
1) the temperature probe of the sensor in the test position fixed safety;And put the cable does not affect the location of the measurement and ensure patient safety;
2) if can't accurately monitoring temperature sensor, it is the location of the wrong place safe or not, if you have any of the above, reposition the sensor or choose another Type sensor.
3) make sure sensor is fixed, the cable has been installed, and plug have been connected to the multi-parameter monitor, and according to the operating procedure operation after 3 minutes can start reading provide for accurate readings, or on the basis of multi-parameter monitor requirement.
4) light to take light put, don't pull a contact between the wire and plug, or wire and contact between the sensing element;In the esophagus and rectum before use, the user must confirm the use of sensors, flexible to apply.Maintenance, cleaning, disinfection and sterilization:
A) check whether the product is intact and cleaned before use;
B) cleaning: clean with a mild detergent or soap solution leaching temperature of cloth.To be dry after cleaning products.
C) disinfection: use chemical disinfectant, such as alcohol, propyl alcohol or phenol disinfectants.Note: do not immerse the product in the liquid.
D) check products, such as found signs of damage or deterioration, don't use it.
Storage method: after cleaning, disinfection, drying products, put it in a sealed plastic bag, inner environment should meet the following conditions: temperature: - 10 ~ + 40 ℃ relative humidity: 0% ~ 80%, atmospheric pressure 86 kpa0 ~ 106 kpa
The warranty service: if you would like to get any after-sale guarantee service information about this product, please contact the local agent.
The data are under preparing to be complete, kindly understand.
For more details, please feel free to contact us, thanks.