Product Name: Neonate Disposable Two Tube NIBP Cuff
Arm circumference: 6.9-11.7 cm;
Tube type: two-tube;
Compatible Models: kinds monitorings.For GE Corometrics, DINAMAP and MDE Monitors, For GE Eagle, Dash, Tram, Solar, MAC-Lab Systems.
Product Structure: consists of tracheal joints, trachea, airbags, sleeves;
(1) Soft TPU (polyurethane):compared to the traditional material, it can filling more gas filling;
(2) Reusable & Convenient: can be used after cleaning and disinfection easily;
(3) Standard Remark:universal signal range and index line enable the choose appropriatly;
(4) Comfatable:additional grinding operation posted increases loose sence for testing;
(5）No letex: not contain latex;
(6) Approved by the ISO 13485 and CE certificated.
Leaking Pressure: for adults and children balloon inflatable cuff to 250 mmHg, there are no more than 4 mmHg/min; for infants and neonates balloon inflatable cuff to 150 mmHg the leaking air pressure no more than 4 mmHg/min.Application Scope: multi-parameter monitor or after device matching, manual measuring blood pressure and non-invasive blood pressure automatic monitoring.
1)Check before use: the compatibility with devices, the characteristic, the performance whether damage. Only complete products can be used.
2)Select the suitable cuff according to the patient’s age, testing area.
3)Check whether the connection of transparent plastic closed or the sensor plug leck.
4)Not to use in MRI scans,the current conductor can cause burns and the interference signal.
5)Elastic cuffs should be wrapped properly in the patient's body to leave about two fingers lose the width of cuff.
Maintenance,Cleaning,Disinfection and Sterilization:
1)Check whether the product is intact and cleaned before use.
2)Cleaning:clean with a mild detergent or soap leaching towel, product need dry after cleaning.
3)Disinfection:use chemical disinfectant, such as alcohol, propyl alcohol or phenol disinfectants.
4)Sterilization : unsing the highest temperature 50 / 122 F to sterilize and then ventilate with Oxygen according to the manufacturer's recommendations.5)Not immerse the product in the liquid, not to expose to the strong ultraviolet radiation.
The data are under preparing to be complete, kindly understand.For more detail, please feel free to contact us, thanks.