Using the special conductor, output with a outstanding data, high shielding, antistatic and low noise, high bending strength transmission, which can persistently used in a high superior anti-aging performance.
Doesn’t contain latex. Outside can be made of the medical grade PVC, TPR, TPE, TPU, PTFE, silicone.
The connection between the probe is also very important to the test results, the position of the cushion is after calibrated and tested, to ensure that the correct location of the transmitter and accurate detection of the receiver.
SpO2 Test Range: 70% ~ 100%;
SpO2 Accuracy Test: 90% ~ 100% ± 2%, 70% ~ 89% ± 3%;
Pulse Rate Test Range: 30 ~ 245 BPM;
Pulse Rate Test Accuracy: ± 3 BPM.
Used for pulse oximetry and multi-parameter monitor, continuous and non-invasive acquisition and transfer of patients (> 40 kg weight adults) to reflect pulse and oxygen saturation signal;
Unsuitable for weak perfusion status and movement state monitoring, the probe is unsuitable for long-term monitoring, should be inspected once every 4 hours or change other finger to measure.
This product may not to be fixed on the position of patients with tissue damage area;
Don’t apply to the patients and users allergic to the PVC, TPU, TPE, ABS plastic.
1)Check before use: the compatibility with devices, the characteristic, the performance whether damage. Only complete products can be used.
2)Connect the sensor to the end of the connecting cable with transparent plastic latch.
3)The transparent plastic latch and sensor plug closed.
4)Connect the end plug adapter cable instrument to the monitor blood oxygen (SpO2) interface.
5)And then according to the instructions on the use of the sensor,fixed the sensor on the patient.
Maintenance,Cleaning,Disinfection and Sterilization:
A)Check whether the product is intact and cleaned before use.
B)Cleaning:clean with a mild detergent or soap leaching towel, product need dry after cleaning.
C)Disinfection:use chemical disinfectant, such as alcohol, propyl alcohol or phenol disinfectants.
Note: do not immerse the product in the liquid, not to expose to the strong ultraviolet radiation.
Working Life:replace every half a year as recommended.
If you still need to use after half year, please evaluate, and inspection should be tightened during the use.
Storage Method: after cleaning, disinfection, drying, put it in a sealed plastic bag.
Inner environment should meet the following conditions: temperature: - 10 ~ + 40 ℃
Relative humidity:no more than 80%, atmospheric pressure 86 kpa~ 106 kpa.
Warranty Service: if you would like to get any after-sale guarantee service information about this product, please contact the local agent.
The data are under preparing to be complete, kindly understand.For more detail, please feel free to contact us, thanks.